If you are unable to find the answer to your query here, please let us know. Our expert development and sales teams will be able to answer your questions directly, and we are more than happy to listen to your views and take on any suggestions concerning the information we present on our website.
Our support telephone numbers are:
U.S. 800-686-4407 (Toll free)
Europe +44 (0)1223 727123
Rest of world (0)1223 727123
or contact us on firstname.lastname@example.org
Yes, you can. The package directly links to Excel, automatically launching the program and entering the results on a worksheet. It is also possible to export GeneTools data as csv (comma separated variable) files which can be utilised with many software packages.
If you select Track Borders it takes a single pixel line down the side of the track and subtracts this pixel for pixel. You do, however, have to be careful that you do not have the edges of the track touching any bands as this can affect the result. Another possibility, avoiding problems caused by the track touching the bands, is to perform a combination of the pixel for pixel subtraction on the borders and the Lowest Slope. This is called Track Borders & Slope subtraction.
A feature of GeneTools is the Rolling Disk background subtraction. The user selects a size of disc (defaults to 30 pixels) which rolls along the track histogram and very closely follows the natural background of the gel.
It is possible to analyse distorted gels. However, to get the most accurate results, you need more than one molecular weight standard. The software will automatically interpolate between the standards as you enter them, trying to follow any distortions within the gel. The more standard lanes you incorporate into the gel, the more accurate the analysis.
The software will automatically detect and compensate for distorted bands. This is achieved by drawing the band marker line to the same angle as the band, thereby correcting the grimace. There is no need for any time-consuming manual adjustments. The software even detects that grimacing of bands often increases in severity with increasing distance from the well position.
Yes. The software will interpolate between them ensuring accurate results even on very distorted or smiley gels.
The software runs a separation technique when analysing a colony image. It counts the colonies initially found, then runs a separation algorithm before recounting them for the final result.
Yes, you can. Just set the relevant parameters in the sample properties dialog box.
There is a facility for overlaying track profiles from different gels for comparison. There is no limit to the number of tracks and gels you can compare at any one time. You can also zoom in and print specific areas of interest within the comparison.
You can import TIFF and BMP images into GeneTools for analysis. However, to maintain Good Laboratory Practice, the results table will include the line 'GLP Warning: This image was created from a non GLP source’.
The analysis software will allow more than one standard to be set on a gel. By selecting the propagation technique, you can accurately analyse even very unevenly run gels. In this case, it is better to use two standards and then select “Use Nearest Standard” option. This will allow for distortions in the local area of the standard. An even more accurate way would be to use three standards and interpolate between them.
Fundamentally, by obtaining the best image prior to analysis. It is most important to frame your sample properly, adjusting the zoom so that it fills as much of the screen as possible. Also the use of integration, either by Auto-exposure or manually, will brighten the image, making analysis easier. When using Auto-exposure do not worry about saturation of bands, the software will prevent this. The bigger and better the image you grab directly from the camera, the easier it is for GeneTools to perform an analysis quickly and automatically. If you obtain a good image, analysis can be carried out in as little as 10 seconds, with the touch of a single button.
Copyright © 2009-2016 Syngene, A Division of Synoptics Ltd